The objective of this proposal is to develop a practical flat-panel x-ray imaging detector with programmable gain in order to address the increasing demand for wide dynamic range flat panel detectors in advanced x-ray imaging applications. The proposed detector employs three major components: a structured cesium iodide (CsI) scintillator to convert x-rays to optical photons;an avalanche amorphous selenium (a-Se) photoconductor, HARP (High-gain Avalanche Rushing amorphous Photoconductor), to convert the optical image to charge and provide a programmable gain;and a large area active matrix (AM) thin film transistor (TFT) array to read out the image electronically in real-time. The proposed detector has been named SHARP-AMFPI (Scintillator-HARP Active Matrix Flat-Panel Imager). It is capable of producing x-ray quantum noise limited images at the lowest dose expected for x-ray imaging (0.1 In the proposed work we will develop the next generation x-ray flat-panel detectors for low dose imaging. It will increase the efficiency of x-ray detection in fluoroscopy by up to 5 times while maintaining the capability for dual mode fluoroscopy/radiography operation by virtue of programmable gain.
Public Health Relevance
In the proposed work we will develop the next generation x-ray flat-panel detectors for low dose imaging. It will increase the efficiency of x-ray detection in fluoroscopy by up to 5 times while maintaining the capability for dual mode fluoroscopy/radiography operation by virtue of programmable gain.
|Scheuermann, James R; Howansky, Adrian; Hansroul, Marc et al. (2017) Toward Scintillator High-Gain Avalanche Rushing Photoconductor Active Matrix Flat Panel Imager (SHARP-AMFPI): Initial fabrication and characterization. Med Phys :|
|Howansky, Adrian; Peng, Boyu; Lubinsky, Anthony R et al. (2017) Deriving depth-dependent light escape efficiency and optical Swank factor from measured pulse height spectra of scintillators. Med Phys 44:847-860|
|Robert, Normand; Watt, Kristina N; Rochette, Sophie et al. (2015) Clinical apparatus for the reduction of dose area product for patients undergoing x-ray catheterization. Med Phys 42:521-30|
|Scheuermann, James R; Goldan, Amir H; Tousignant, Olivier et al. (2015) Development of solid-state avalanche amorphous selenium for medical imaging. Med Phys 42:1223-6|
|Hu, Yue-Houng; Zhao, Wei (2014) The effect of amorphous selenium detector thickness on dual-energy digital breast imaging. Med Phys 41:111904|
|Xiang, Liangzhong; Han, Bin; Carpenter, Colin et al. (2013) X-ray acoustic computed tomography with pulsed x-ray beam from a medical linear accelerator. Med Phys 40:010701|
|Goldan, A H; Zhao, W (2013) A field-shaping multi-well avalanche detector for direct conversion amorphous selenium. Med Phys 40:010702|
|Bubon, Oleksandr; Decrescenzo, Giovanni; Zhao, Wei et al. (2012) Electroded avalanche amorphous selenium (a-Se) photosensor. Curr Appl Phys 12:983-988|
|Wronski, M; Zhao, W; Tanioka, K et al. (2012) Scintillator high-gain avalanche rushing photoconductor active-matrix flat panel imager: zero-spatial frequency x-ray imaging properties of the solid-state SHARP sensor structure. Med Phys 39:7102-9|
|Hu, Yue-Houng; Zhao, Wei (2011) The effect of angular dose distribution on the detection of microcalcifications in digital breast tomosynthesis. Med Phys 38:2455-66|
Showing the most recent 10 out of 25 publications